This Guide is designed to help you to determine the Fluorescence Quantum Yields (QY) of liquid samples using your FLS920 spectrometer and the Integrating Sphere accessory.
Fluorescence Quantum Yields (QY) measurements can be calculated using the relative method by comparing fluorescence parameters to a published quantum yield standard, or they can be determined by the absolute method in which the number of absorbed photons of a sample and the number of consequently emitted photons are measured. This guide explains how to use the sphere by means of the Absolute Quantum Yield method.
Specific measurement parameters are required to accurately and reliably perform quantum yield measurements. In instruments specifically designed with the sole purpose of Quantum Yield (QY) measurement these parameters are invariably fixed. The FLS920 is a flexible, research grade fluorescence spectrometer where the integrating sphere accessory for the measurement of fluorescence quantum yields is only one of its many applications. It is therefore very important that the measurement parameters of the FLS920 are fixed and consistent when QY measurements are being made.
Accurate and reliable Quantum Yield measurements require practice. They also can be quite time consuming. It is not only a question of having the proper instrumentation and having it correctly setup, but also a question of following the correct procedure.
There are two instrumental factors that can have dramatic effects on the fluorescence quantum yields measurements:
Liquid Samples are measured in standard (10mm x 10mm) quartz cuvettes with PTFE (white) stopper. You need ideally 2 identical cuvettes, one containing the sample, the other containing the solvent only (to be used as a blank). When preparing the cuvettes ensure that both sample and blank have the same filling height. 3ml of sample volume is ideal. Do not use less than 2ml.
You also need an absorption (UV-VIS) spectrometer (or the EI absorption accessory) to set up absorbance (OD) values of your sample.
The FLS920 Fluorescence Spectrometer is now discontinued. However, we still fully support all upgrades, parts and replacements, servicing and maintenance for existing FLS920 systems and customers. The FLS1000 Photoluminescence Spectrometer has replaced the FLS920, and is able to measure fluorescence quantum yields, and much more. If you would like more information on upgrades to your existing FLS920 system, or would like to enquire about an FLS1000, simply contact a member of our sales team at firstname.lastname@example.org.
If you have enjoyed this Technical Note, and want to be the first to find out about our research work, applications and product information, then why not sign-up to our infrequent newsletter via the red sign-up button below, or follow us on social media.